ASTM D determines the adhesion strength of rubber to rigid materials, such as metals. The standard requires that the testing machine conform to practices. This standard is issued under the fixed designation D ; the 1 These test methods are under the jurisdiction of ASTM Committee D11 on. This standard is issued under the fixed designation D ; the number These test methods are under the jurisdiction of ASTM Committee D11 on For.
|Published (Last):||2 October 2012|
|PDF File Size:||1.97 Mb|
|ePub File Size:||17.87 Mb|
|Price:||Free* [*Free Regsitration Required]|
Each puck should be. This test is almost identical to Method B, except it applies exclusively to the rubber lining of a tank wall. Please confirm that you agree with our privacy and cookies policy to submit this form. This astk will be used to clamp in the jaws of the grip. Method G of ASTM D measures the bond strength of rubber to metal components with a double shear cylindrical specimen.
This test combines tensile and shear forces in the rubber specimen material. Some of the tests are aimed at shear strength of that adhesion interface. This test is often used to find which adhesive is best for the given rubber.
Please confirm that you agree with our privacy and cookies policy to submit this form. Bluehill Asfm Software is built from the ground-up for touch interaction and an intuitive user experience. They are designed with standard and optional features that increase testing efficiency and improve the testing experience for the operator.
ASTM D429 Rubber Property – Adhesion to Rigid Substrates
The specimen for this method should be a strip of rubber 6. The assembly is cured in a degree Celcius oven for one hour. Loads as high at 10, pounds can be common. How can we help you? Concrete, Asphalt and Rock.
Bluehill Universal Brochure Bluehill Universal Software is built from the ground-up for touch interaction and an intuitive user experience.
It may apply to vibratory or vibration applications, although this test is not dynamic. The sample also lends itself to testing the effectiveness of different processing techniques and various adhesive systems.
Before the load is applied, strip the rubber from the metal plate by using a sharp knife. In the Figure 7 diagram, the specimen is shown in a horizontal position, when in reality, the specimen would be placed vertically in the test machine.
Once the test is complete, you will want to record the force vs displacement. Test Method A helps determine the static adhesion strength of rubber to metal. Peel fixtures provide a condition of use to measure the bond strength between laminated surface coatings, films, packaging, and their rigid backings, as well as other bonded structures. Email addresses must contain the symbol. Our Series and Series universal testing systems meet this specification and provide an appropriate solution for this test application.
Method B describes a peel procedure to determine the adhesive strength between rubber materials to a metal plate or substrate. Place the tab in the grip and apply a steady load at a test speed of 50 mm 2.
ASTM D Rubber Property – Adhesion to Rigid Substrates – Instron
This test standard is divided into methods A through H. Next Checking to see if you are a returning visitor…. Repair Services and Parts. This ensures that the line of tensile force remains ast with the grip body. Check this box to receive pertinent commercial email from Instron satm on products, upcoming events, and more!
Make sure that a 1 inch by 1 inch tab is kept from bonding by covering it with tape.